Service Manuals, User Guides, Schematic Diagrams or docs for : HP 5962-8119<< Back
| HomeMost service manuals and schematics are PDF files, so You will need Adobre Acrobat Reader to view : Acrobat Download
Some of the files are DjVu format. Readers and resources available here : DjVu Resources
For the compressed files, most common are zip and rar. Please, extract files with Your favorite compression software ( WinZip, WinRAR ... ) before viewing.
If a document has multiple parts, You should download all, before extracting.
Good luck. Repair on Your own risk. Make sure You know what You are doing.
Image preview - the first page of the document
>> Download 5962-8119 documenatation <<Text preview - extract from the document
Agilent Model 66332A
Dynamic Measurement DC Source
and Agilent Model 6632B, 6633B, 6634B
System DC Power Supply
Agilent Part No. 5962-8119 Printed in Malaysia
Microfiche No 6962-8120 October, 2003
Agilent Technologies certifies that this product met its published specifications at time of shipment from the factory.
Agilent Technologies further certifies that its calibration measurements are traceable to the United States National
Bureau of Standards, to the extent allowed by the Bureau's calibration facility, and to the calibration facilities of other
International Standards Organization members.
This Agilent Technologies hardware product is warranted against defects in material and workmanship for a period
of one year from date of delivery. Agilent Technologies software and firmware products, which are designated by
Agilent Technologies for use with a hardware product and when properly installed on that hardware product, are
warranted not to fail to execute their programming instructions due to defects in material and workmanship for a
period of 90 days from date of delivery. During the warranty period Agilent Technologies will, at its option, either
repair or replace products which prove to be defective. Agilent Technologies does not warrant that the operation for
the software firmware, or hardware shall be uninterrupted or error free.
For warranty service, with the exception of warranty options, this product must be returned to a service facility
designated by Agilent Technologies. Customer shall prepay shipping charges by (and shall pay all duty and taxes)
for products returned to Agilent Technologies. for warranty service. Except for products returned to Customer from
another country, Agilent Technologies shall pay for return of products to Customer.
Warranty services outside the country of initial purchase are included in Agilent Technologies' product price, only if
Customer pays Agilent Technologies international prices (defined as destination local currency price, or U.S. or
Geneva Export price).
If Agilent Technologies is unable, within a reasonable time to repair or replace any product to condition as warranted,
the Customer shall be entitled to a refund of the purchase price upon return of the product to Agilent Technologies.
LIMITATION OF WARRANTY
The foregoing warranty shall not apply to defects resulting from improper or inadequate maintenance by the
Customer, Customer-supplied software or interfacing, unauthorized modification or misuse, operation outside of the
environmental specifications for the product, or improper site preparation and maintenance. NO OTHER
WARRANTY IS EXPRESSED OR IMPLIED. AGILENT TECHNOLOGIES. SPECIFICALLY DISCLAIMS THE
IMPLIED WARRANTIES OF MERCHANTABILITY AND FITNESS FOR A PARTICULAR PURPOSE.
THE REMEDIES PROVIDED HEREIN ARE THE CUSTOMER'S SOLE AND EXCLUSIVE REMEDIES. AGILENT
TECHNOLOGIES SHALL NOT BE LIABLE FOR ANY DIRECT, INDIRECT, SPECIAL, INCIDENTAL, OR
CONSEQUENTIAL DAMAGES, WHETHER BASED ON CONTRACT, TORT, OR ANY OTHER LEGAL THEORY.
The above statements apply only to the standard product warranty. Warranty options, extended support contacts,
product maintenance agreements and customer assistance agreements are also available. Contact your nearest
Agilent Technologies Sales and Service office for further information on Agilent Technologies' full line of Support
The following general safety precautions must be observed during all phases of operation of this instrument. Failure to comply
with these precautions or with specific warnings elsewhere in this manual violates safety standards of design, manufacture, and
intended use of the instrument. Agilent Technologies assumes no liability for the customer's failure to comply with these
Servicing instructions are for use by service-trained personnel. To avoid dangerous electrical shock, do not perform any servicing
unless you are qualified to do so. Some procedures described in this manual are performed with power supplied to the instrument
while its protective covers are removed. If contacted, the energy available at many points may result in personal injury.
BEFORE APPLYING POWER.
Verify that the product is set to match the available line voltage, the correct line fuse is installed, and all safety precautions (see
following warnings) are taken. In addition, note the instrument's external markings described under "Safety Symbols"
GROUND THE INSTRUMENT.
Before switching on the instrument, the protective earth terminal of the instrument must be connected to the protective conductor
of the (mains) power cord. The mains plug shall be inserted only in an outlet socket that is provided with a protective earth
contact. This protective action must not be negated by the use of an extension cord (power cable) that is without a protective
conductor (grounding). Any interruption of the protective (grounding) conductor or disconnection of the protective earth
terminal will cause a potential shock hazard that could result in personal injury.
Only fuses with the required rated current, voltage, and specified type (normal blow, time delay, etc.) should be used. Do not use
repaired fuses or short-circuited fuseholders. To do so could cause a shock or fire hazard.
KEEP AWAY FROM LIVE CIRCUITS.
Operating personnel must not remove instrument covers. Component replacement and internal adjustments must be made by
qualified service personnel. Do not replace components with power cable connected. Under certain conditions, dangerous
voltages may exist even with the power cable removed. To avoid injuries, always disconnect power, discharge circuits and
remove external voltage sources before touching components.
DO NOT SERVICE OR ADJUST ALONE.
Do not attempt internal service or adjustment unless another person, capable of rendering first aid and resuscitation, is present.
Any adjustment, maintenance, and repair of this instrument while it is opened and under voltage should be avoided as much as
possible. When this is unavoidable, such adjustment, maintenance, and repair should be carried out only by a skilled person who
is aware of the hazard involved.
DO NOT SUBSTITUTE PARTS OR MODIFY INSTRUMENT.
Because of the danger of introducing additional hazards, do not install substitute parts or perform any unauthorized modification
to the instrument. Return the instrument to an Agilent Technologies Sales and Service Office for service and repair to ensure that
safety features are maintained.
Refer to the table on the following page
WARNING The WARNING sign denotes a hazard. It calls attention to a procedure, practice, or the like, which, if not
correctly performed or adhered to, could result in personal injury. Do not proceed beyond a WARNING sign
until the indicated conditions are fully understood and met.
Caution The CAUTION sign denotes a hazard. It calls attention to an operating procedure, or the like, which, if not
correctly performed or adhered to, could result in damage to or destruction of part or all of the product. Do
not proceed beyond a CAUTION sign until the indicated conditions are fully understood and met.
Safety Symbol Definitions
Both direct and alternating current
Three-phase alternating current
Earth (ground) terminal
Protective earth (ground) terminal
Frame or chassis terminal
Terminal is at earth potential (Used for measurement and control circuits designed to be
operated with one terminal at earth potential.)
Terminal for Neutral conductor on permanently installed equipment
Terminal for Line conductor on permanently installed equipment
Units with this symbol are not completely disconnected from ac mains when this switch
is off. To completely disconnect the unit from ac mains, either disconnect the power
cord or have a qualified electrician install an external switch.
In position of a bi-stable push control
Out position of a bi-stable push control
Caution, risk of electric shock
Caution, hot surface
Caution (refer to accompanying documents)
The information contained in this document is subject to change without notice. Agilent Technologies makes no
warranty of any kind with regard to this material, including but not limited to, the implied warranties of
merchantability, and fitness for a particular purpose.
Agilent Technologies shall not be liable for errors contained herein or for incidental or consequential damages in
connection with the furnishing, performance or use of this material.
This document contains proprietary information which is protected by copyright. All rights are reserved. No part of
this document may be photocopied, reproduced, or translated into another language without the prior written
consent of Agilent Technologies, Inc.
◦ Jabse Service Manual Search 2019 ◦ Jabse Pravopis ◦ Other service manual resources online : Fixya ◦ eServiceinfo